It can be positioned exactly on colour and paper charts for accurate hand measurements. Phoenix, Arizona , Ships to: In fact, they are no longer around because they were purchased and taken over by X-Rite many years ago, who also no longer supports the SpectroScan. This is not the most elegant way to fix a problem. Each product is tested and is in good working condition when it is shipped Unless it is marked for parts or repair. We strive to make our customers happy and want you to feel that we have treated you with the ultimate respect.
|Date Added:||13 November 2012|
|File Size:||11.77 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
In order to return the item, you must email us first and obtain an RMA number for us to process. I also thought about the original error code: This page was last modified on 3 Augustat That seems to leave just the right amount of dry lubricant to the rail surface to provide easy gliding.
To get access to the rails you will need to remove the cover off the upper unit spectrkscan moves forward and back across the top of the table.
Using a multimeter, I measured Software for Konica Minolta Instruments. Therefore, I was beginning to hypothesize that I had either a power or communications issue between the main circuit board and the SpectroLino. The diagram below from the User Guide shows the setup. The Spectrolino has outstanding technical performance.
▷ Used Spectral photometer Gretag Macbeth SpectroScan mit Spectrolino for sale –
At this point I began thinking that the issue was with the SpectroScan table and not the SpectroLino, but I wanted confirmation that my SpectroLino was working. Most of the time, people are impressed with the geetagmacbeth we do because we customize our own skids for larger items and we take the utmost care to provide a safe and cozy journey for your item. Although polite, the individual I was conversing with claimed that X-Rite no longer supported or serviced these units and that a schematic or service manual was unavailable.
It was time to re-install the board into the SpectroScan and test. We have seen even a single hair on a rail to cause the movement to stick, so don’t underestimate the importance of cleaning the rails.
I immediately rgetagmacbeth the problem. After choosing the correct Test ChartI click the Start button to begin the process. There are more rails inside.
The front control panel circuit board mounts underneath the top surface of the SpectroScan table: Although I had previously measured 24v on pin 7, I surprisingly discovered that I had no continuity on this pin.
Validation of Bank connection. We do not except returns that have been damaged through shipping without the buyer following the proper procedure, if the customer receives an item that appears damaged from the shipping carrier that looks mishandled, damaged or heavily handled by the appearance of the box we strongly recommend our customer to refuse the shipment and notify us immediately.
Want to really stop the chat history? X-Rite and similar handhelds. Either way, I was unaware of any repair company to help me, so I was forced to attempt the repair myself. Sending something back without first receiving a valid RMA number will exclude you from receiving credit.
Colour samples, paper samples, transparencies, colour charts and monitors.
GRETAGMACBETH SPECTROSCAN WHITE Densitometer Spectrolino – $ | PicClick
This is no longer the case as a new driver is now available for Windows 7, so I can install ProfileMaker on just about any computer. Color eXchange File format.
Food and Drug Administration and state and local regulatory agencies. Commands to move a particular motor are first sent to the main processor, no matter if they emanate from the serial line or the front panel.
Immediately upon power up, I knew I had fixed the SpectroScan when a series of beeps were gretagmmacbeth from the SpectroLino and it moved itself over the calibration tile. Contents 1 Disclaimer 2 Maintenance for SpectroScan tables 2. With this information, I knew each pins designation, which allowed me to measure for power at the mini-DIN connector that the SpectroLino plugs into.
Indeed, I actually spoke with him many years ago regarding another issue with my SpectroScan. I am a dealer.